When is Calibration required?
After performing the calibration the device will last a full vaccination event.
We also recommend performing the calibration in the following cases to maximize your Injection event results-
At the beginning of the day.
After replacing an Injector.
If you notice air bubbles in the vaccine tubes.
How to perform Calibration?
-Please notice- From X.4.11 The Calibration is configured at Normal speed.
Step by Step
Turn on the Control Unit
2. Navigate to the Calibration screen.
3. Click on the OK button to start the Calibration process.
4. Take a syringe in your hands and hold it facing up or use a measuring canister.
With an automatic Injection Head- Press the trigger on the Hand Unit and pull
back the Injection Head.
With a manual Injection Head, just press the trigger.
The system will automatically deliver ten doses.
5. Check the injected material volume in the syringe.
If the dosage in the syringe is the correct amount, press the down arrow to
reach the Save field, click OK, and continue to step 6.
If the dosage in the syringe is not correct, go to the 'Volume' field and click OK.
When you see the full arrow, use the up and down arrows to set the volume
that is currently in the syringe.
When done, Click OK.
Go to the Save field and click OK.
6. Press OK to stop the Calibration mode.
When using a Dual system, after calibrating the first vaccine bottle, repeat the process for the second vaccine bottle.